Attention point of maintenance of Laparoscopic Surgical Instruments
The instrument should be dry as it is assembled, check the needle on the rubber cap, a rubber valve is damaged, should be replaced if damaged, there is a place with oil, and check whether the flexible coupling, the bite is close, screw loosening, lack of timely maintenance, in order to supplement. The equipment on the lubricating oil should dry the excess oil so as to avoid the effect of excessive lubricating oil on the disinfection effect. The maintenance of equipment should be according to a certain order into the instrument box code neatly, and count the correct sterilization indicator card, double cloth paste packing people indicator tape, and indicate the item name and the effective date of disinfection.
Zhejiang Geyi Medical Instrument Co., Ltd.
Address: No. 190 Chutian Road, Xixing Street, Binjiang Zone, Hangzhou, Zhejiang, China
Contact: Desiree Shu
Mobile: +86 15868407021